The Panametrics 25DL PLUS ultrasonic precision thickness gage provides cost-effective solutions in applications where the opposite side of the test material is difficult or impossible to reach. An important benefit of this handheld unit is its ability to make thickness measurements on many materials in a wide variety of shapes and sizes. A large Liquid Crystal Display displays both the live ultrasound waveform and the thickness reading. Panametrics Model 25dl Operators Manual Operation of this handheld gage is intuitive with a color-coded keypad and direct-access number keys. One of the many unique features of this gage, Application Auto-Recall, allows the user to select and recall any of the stored standard or custom transducer setups in the gage's memory.
Select Which transducer you are going to use.
You may need to experiment with each probe to determine which is the most appropriate. To configure the unit you will need two test pieces of the same material which you want to test that each have known thicknesses. One thick and one thin. CAL Setup. Use the Up and Down arrow keys and select the transducer which you have decided to use. Search this site. Navigation TechRentals.
Training Information. Product Notes.
the model 25DL. Crescent Street. Waltham, MA USA. () Toll Free USA () Fax () http://www. panametrics.
Product Summaries. Industry Lists. Three are supplied with the kit. The unit will turn on.
A list of transducers will appear. If you are using the M transducer hold the transducer in the air and press Zero briefly. Place couplant on the thick material. Hold the transducer to the thick test piece Wait for a stable reading. Enter the known thickness of the thick test piece using the keypad.Using contact transducers. Mode 3: Time interval between successive backwall echoes following the first interface echo after the excitation pulse. All specifications are subject to change for product improvement without notice. Mode 2: Time interval between the first interface echo after the excitation pulse and the first backwall echo. Using delay line or immersion transducers.
Place couplant on the thin material. Hold the transducer to the thin test piece Wait for a stable reading.
This will save the calibration figure.